Gostritzer Str. 61 - 63
01217 Dresden, de
+49 (351) 871-8610
High-end Camera Series ImageIR®
Thermography systems for highest demands
The optical channel consists of interchangeable infrared lenses as well as apertures, filters, reference and optical elements for use in different applications. High-precision and robust fine mechanics combined with controlled actuators and sensors enable uncompromising radiometric stability under all application conditions. A well-balanced range of interchangeable precise radiometric infrared lenses in measurement quality covers the fields of telephoto lenses, standard lenses, wide-angle lenses as well as macroscope and microscope lenses.
Cooled focal-plane-array photon detectors of different types (InSb, MCT and QWIP) and of different formats (320 x 256) and (640 x 512) IR-pixels working in snapshot mode are used. Depending on the detector, the frame rates can reach some hundreds of Hz in full screen mode as well as some ten thousands of Hz in part screen mode. The integration times range between 1 µs and 2 ms – at a measurement accuracy of +/- 1 K, or 1 %. The average temperature definition is better than 0.02 K at an object temperature of 30 °C.
Depending on the chosen interface option, the ImageIR® thermography systems contain flexibly configurable digital interfaces for the transfer of image data and for controlling, such as Gigabit Ethernet, Camera Link, USB and CAN. An optional trigger interface guarantees most precise synchronicity from and to external sensors and actuators, which, is required by active thermography applications, for example.
Combined with the thermography evaluation software IRBIS® 3 professional by InfraTec, the camera series ImageIR® sets completely new standards as an efficient high-performance tool for most challenging thermographical testing.
Die Nutzung von hier veröffentlichten Informationen zur Eigeninformation und redaktionellen Weiterverarbeitung ist in der Regel kostenfrei. Bitte klären Sie vor einer Weiterverwendung urheberrechtliche Fragen mit dem angegebenen Herausgeber. Bei Veröffentlichung senden Sie bitte ein Belegexemplar an email@example.com.